The operations manual should be consulted in all cases where this symbol is marked. It must be consulted in all cases in which the symbol is marked, to know the nature of the potential HAZARDS and any action to be taken to avoid them.
- Caution, possibility electric shock
The incorrect operation can cause electric shock or damage to the instrument.
- Warning; Hot surface
The incorrect operation can cause burns.
The suggestion includes various operating tips and reasonable methods for setting up and using the instrument.
- Environmental Protection use period
The EPUB (Environmental Protection Use Period) symbol is prepared in accordance with the Measures for the Control of Contamination of Electronic Information Products and Marking for the Control of Pollution Caused
by Electronic Information Products (SJ / T11363-2006) and is suitable for electronic information products
sold in China.
The reagent used in the position with the symbol attached to the instrument may result in a biohazard, and therefore, the protection will suffer.
- Protective conductor terminal
The position with the symbol attached to the instrument indicates the ground terminal.
The following basic safety measures should be observed during the operation, maintenance and repair of the instrument. If the notes, warnings or cautions specified in the manual are neglected, the basic protection provided by the instrument may be affected. Meanwhile, it may also damage the security level of the design and manufacture of instruments and the intended scope of application.
It is not allowed to change the parts of the instrument without permission, and the inspection and maintenance of the instrument must be completed by engineers licensed by Hangzhou AnYu Technologies Co., Ltd. If the equipment is not used in the manner specified in this manual, the protection provided by the equipment may be damaged
The wells and hot cap of the instrument should be periodically cleaned with special brushes bathed in a little absolute alcohol to guarantee complete contact between tubes and wells and high thermal conductivity while free
of pollution. Use a soft cloth to do daily maintenance to clean the surface and wells. If there is any stain on the surface, you can clean it with a dust-free cloth with 75% alcohol. Do not use chlorine disinfectants, moisturizers, or antibacterials soaps, aromatic solutions, propanol or isopropanol to clean the equipment, as they are types of cleaning or sanitizing agent is easy to have a chemical reaction with parts of the equipment or materials contained in the equipment and cause danger.
Perform only the maintenance mentioned in this manual. Before using any cleaning and disinfection method not described in this document, please contact the technical support personnel of the Company or authorized agency to ensure that the method used will not damage the instrument. If you have any questions on the compatibility of disinfectants or cleaning agents with pieces of equipment or materials contained in the equipment, consult the Company or an authorized agency.
Main Applications and Features of the AGS8830-8, AGS8830-16 Real-Time PCR Detection System
The product is based on the fluorescent polymerase chain reaction (PCR) and is used in conjunction with supportive nucleic acid detection reagents for Quantitative and qualitative detection of nucleic acid (RNA / DNA) samples from
the human body, including pathogens and human gene projects. Attention: When the instrument is used in clinical detection, the reagent must be within the validity period and in accordance with local laws and regulations. Positive quality samples and negative and critical quality samples should be tested in each experiment, when the quality control samples meet the quality control requirement, the experiment is efficient.
The instrument is characterized as follows:
- Advanced semiconductor cooling technology is used to control temperature accurately;
- 7-inch LCD screen, user-friendly operation interface, simple operation;
- With the power-off protection function, continue to complete the unfinished procedures after the resumption of power supply;
- USB 2.0 interface supports U disk to store data;
- Large data storage and can store more than 1000 files.